Manufacturer: Dentsply Sirona
The ProPex Pixi Apex Locator is another innovative solution from Dentsply Sirona & Maillefer, encouraging the standard of care in Endodontics and assisting clinicians to deliver more predictable endodontic outcomes.
Propex Pixi is not recommended for use:
Propex Pixi™ is an electronic device used for apex location during root canal treatment. Propex Pixi™ must only be used in hospital environments, clinics or dental offices, by qualified practitioners.
Contraindications Propex Pixi™ is not recommended for use: • In patients who have a pacemaker or other implanted electrical devices, or have been cautioned by their physicians against the use of small electric appliances such as shavers, hair dryers, etc; • In patients allergic to metal; • In children. 3
• The scale indication on the Propex Pixi™ screen does not represent a distinct length or distance in mm or other linear units. It simply indicates the file progression towards the apex;
• The following patient related factors may prevent accurate readings: - Blocked root canals; - Teeth with large apices; - Root fracture or perforation; - Metal crowns or bridges, if they come into contact with the file or the lip clip.
• Inaccurate or incorrect readings due to the environment are likely to occur in the following cases: - Presence of portable or movable radio frequency transmitters in the surroundings; - Film viewers or other Illumination devices which use an inverter may cause abnormal operation of the apex locator. Such devices should be turned off during use of the Propex Pixi™; - Electromagnetic interference could cause improper operation of the device. In such cases the device behaviour may become abnormal or random. Usage of any devices emitting electromagnetic radiation, such as cellular phones, remote controls, transceivers, etc., should be prohibited in the vicinity of Propex Pixi™.
• General safety warnings: - In order to prevent infectious agent transfer it is highly recommended to use a rubber dam system during the endodontic procedure. - Make sure that the lip clip, hook or fork does not come into contact with an electric power source such as an electrical socket. This could result in a severe electrical shock. - Do not use Propex Pixi™ in the presence of flammable substances. • Only use the original charger. • Use AAA 1.2V 1000mAh NiMH rechargeable batteries only. Usage of non rechargeable batteries may cause device damage. Recommended battery type: Manufacturer – GP Batteries
Out of Stock
Be the First to Know. Sign up for upcoming deals and offers today
Shop no. 44 , Block -C, Second Floor , Near Gate No.2, Palam Vihar Vyapar Kendra, Gurgaon, Haryana- 122017, India
Copyright ©2021 ID Enterprises, Inc. All rights reserved.