Mani Peso Reamers Ideal for widening and paralleling root canals for post-core preparations. Hard Fiber Stainless steel. Right angle.
A new material developed by Mani has made it possible to commercialize safe and sharp peso reamers with higher anti-corrosion resistance to any kind to sterilization including autoclave, chemical vapor, and super acid water.
Peso Reamers are used to enlarge the cervical portion of the canal and to prepare the canal entrance.
These instruments are also useful in removing gutta-percha in preparation for creating post space.
In case of absence of a crown part because of dental caries, this instrument can be used as a post for strength retention when building a crown on a root using remaining part.
Products’ bending durability must exceed specified value of our standard.
Remove filling material using MANI PEESO REAMERS after verifying length of cured root canal in X-ray photograph and building an abutment. The limit of depth is two-thirds of root length. Shape root canal with MANI PEESO REAMERS corresponding to root canal diameter. If needed, demineralize inside root canal and tooth surface. Insert this product after kneading filling material and filling up root canal with it. Additionally, put core buildup material on projecting part from root canal and calcify. Take impression and make crown prosthesis after the final abutment building is done.
STEP 1 Formation of post orifice
STEP 2 Selection of EG POST
Size chart 1 MANI PEESO REAMERS Unit: mm
Size chart 2 MANI EG POST Unit : mm
MANI PEESO REAMERS and MANI PASTE CARRIERS must be cleaned or sterilized before using. Also keep the allowable engine speed strictly. **
※ MANI PEESO REAMERS Allowable min-¹.: less than 1200 min-¹.
※ MANI PASTE CARRIERS Allowable min-¹.: less than 800 min-¹.
※ For the details of MANI PEESO REAMERS and MANI PASTE CARRIERS, refer to each instructions for use.
Precautions for use
Storages and duration of use
Maintenances and inspections
Method of sterilization:
Put this product in a sterilization pack(or foil) and place it on a sterilization tray for autoclave sterilization with reference to the following terms .*
Terms for sterilization:
condition(1) temperature: 121degrees Celsius time: 20 minutes or more condition(2) temperature: 126degrees Celsius time: 15 minutes or more
Bd. Général Wahis 53,
※ Always keep this document near at hand. In case of loss, please contact dental department of the manufacturer for a copy.
※ The specifications, structures and materials are subject to be changed without previous notices for the demands of improvement.
※ The contents of this instruction manual are subject to be revised without previous notices.
※ * and **mark indicate revised points.
Be the First to Know. Sign up for upcoming deals and offers today
Coltene, over the years since its establishment, has emerged as one of the le...
Good oral health plays an important role in our lives. It has several life-ch...
Good oral health and hygiene have plentiful benefits. It not just earns you a...
Dental braces are dental devices that orthodontists use to reposition and str...
Dental bonding also known as teeth bonding is a cosmetic procedure done to re...
Shop no. 44 , Block -C, Second Floor , Near Gate No.2, Palam Vihar Vyapar Kendra, Gurgaon, Haryana- 122017, India
Copyright ©2021 ID Enterprises, Inc. All rights reserved. Developed By DO Communication